Class: Instrumentation Manufacturer: Kajaani Model No: SD Part Number: Qty: 1. Description: Automatic Sample Valve. follow-up through a communicator or Kajaani interface via Ethernet. Three operating modes: • Manually initiated. • Collection timer 24/7. STEC SEC Mass Flow Controller. ₪. METSO KAJAANI SD TRANSMIITER USED. ₪. MKS Flow Meter (N2) – Model ALS3BV.
|Country:||Sao Tome and Principe|
|Published (Last):||27 June 2005|
|PDF File Size:||13.40 Mb|
|ePub File Size:||19.70 Mb|
|Price:||Free* [*Free Regsitration Required]|
Sampling device removed from process 4 2 3 1 5 7 6 1. Connection box, front view. Custom Bundle see all Custom Bundle. Refine more Format Format. SD installation, operating, service – A5 – W V1. Removing the seal kkajaani The device may have been in contact with toxic or otherwise hazardous substances.
Sell now – Have one to sell? Sign in to check out Check out as guest. See all industrial has no other items for sale. The samples are diluted with water and transported to the analyzer along a sample line, aided by water pressure. Kauaani the correct sampling device 1.
Installing the closing piston: Neither receipt nor possession thereof confers or transfers any right to reproduce or disclose the document, any part thereof, any kajaanni article or device, except by written permission from the manufacturer. Removing the bearing bush and O-ring. Test the operation of the actuator and ball valve with instrument air.
KURZ 454FTB-16-HHT Single Point Insertion Thermal Mass Flow Transmitter
SD installation, operating, service – 6. Kaajani gap 13 mm b Air inlet to sampling device Amounts shown in italicized text are for items listed in currency other than Canadian dollars and are approximate conversions to Canadian dollars based upon Bloomberg’s conversion rates. Items in search results. Report item – opens in a new window or tab. Hose couplings ssd5002 19 mm and 25 mm hoses are included in delivery.
Calculate Varies based on location and shipping method. Mount the sampling device on a vise, in the position shown in Fig. If the actuator has been detached, fasten it loosely in position, so sv5002 the fastening nuts parts 20 and 21 are at the actuator piston end of the threaded part. Detach the collar part 3. Samples can be taken from all post-digester process stages.
מדי זרימה נוזלי והמונית
Tighten nut 20 properly — make sure that the connection does not move sideways when tightening it! Learn More – opens in a new window or tab.
If the material is not indicated on a component, see the device manuals for further information. No air must leak from the connector. Then pull the bush out of its casing by turning nut A. Pull the seal ring out using the extracting tool in the special tool set.
Format see all Format. Water and air inlet connections of the sampling device. Metso will accept only thoroughly cleaned devices, and the cost of recycling will be invoiced from the customer; please ask for recycling quotation. Water to sampling device Continue from point 6.
Brand see all Brand. Loosen nut 20 slightly to make sure that the actuator is accurately aligned with the closing piston shaft. Disconnect sample hose from sampling device. This item will ship to United Statesbut the seller has not specified shipping options.
The flow opening of the piston opens more when the piston moves a longer distance. Replace the seals indicated in Fig. This item will be shipped through the Global Shipping Program and includes international tracking.
מדי זרימה נוזלי והמונית: פשוט לקנות באיביי בעברית – זיפי
Loosen the locking screw hex socket screw M6, part 7. Turn the shaft 5 by the 13 mm spanner gap to the required direction.
Replacing the actuator diaphragm Add to watch list. Take the seal ring and installation tool from the freezer.